53181A LABVIEW DRIVER
I have also tried to use the provide NI measurement VI and do not get anything from the unit either. General Use Programming and Remote Communications clear selected. I have made a Config VI which I have attached. I’m having difficult in communicating with the frequency counter I’m using. Then I’ve tried to install that add-in manuall Get a feed of this content Use this view in a tile. I have downloaded the driver from here, so I can open your VI now:
|Date Added:||8 October 2016|
|File Size:||69.88 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I need to make successive second 2 hour period measurement and I want them to be gapless. Looking at the x driver, you will see i Did you read the documentation?
I use a A frequency counter to measure frequency of a signal.
Agilent 53181 Counter/Timer
Log in to follow, share, and participate in this community. Hi, Does any one has labview driver for agilent A screen capture using gpib? If you open the “VI Tree”, you can see all the driver VIs, and also their documentation turn on the “Help” question mark. The code runs through fine and shows no errors until it gets to the Initialise VI and more specifically, the Visa Read section where it is looking for something from the equipment.
Hi After update my A from firmware 1. At present, I’m 533181a trying to establish comms but I have VI’s made to log all the data and graph it etc. Then I’ve tried to install that add-in manuall I have also tried to use the provide NI measurement VI and do not get anything from the unit either.
Frequency Counters | Keysight Community
I have downloaded the driver from here, so I can open your VI now: Visa Write shows no errors. The problem is that you mixed up the proper 5311a
You can also try the shipped examples find it via the Example Findersee the last screenshot below! I keep getting a time out error and cant get a readout from my vi.
I used configure voltage driver to fix the voltage. In your VI you run first the “Config” vi, and then the Initialize vi. Hi, I’d appreciate any help in reading frequency data from an HP B. I finally found the manual and tried some Most Active Software Boards: I have made a Config VI which I have attached.
Record frequency automatically from A | Keysight Community
I finally found a manual and tried u One thing I have noticed is that the ‘Remote’ light on the front of the unit never illuminates but I’m not sure if its supposed to. I’m having difficult in labvieew with lahview frequency counter I’m using. Thanks for the reply. So it takes gates every other measure Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Looking at the attached driver project, I’ve attached the old and new driver LV I don’t know if I’m missing something obvious or if the problem lies deeper but any help would be appreciated.